Dental X-ray Film Automatic Processor Developer
The oral automatic X-ray film bath equipment is a total automatic
and wall mounted type.
Easy to clean and maintenance, stable running, small noise,
operators do not touch the developers and fixers directly,
convenient and safe operation.
The film is bathed automatically through three step: Developing,
fixing and bathing.
The time developing films needed can be selected in the designing
During the bathing process, delivery mechanism adopts anti-flaming
ABS engineering materials.
2. Main Technical parameters
Normal working conditions
A). Surroundings temperature: 18o C-30o C;
B). Relative humidity:≤70%;
C). Atmospheric pressure: 700hPa-1060hPa;
D). Voltage of electrical source:110V~ 220V± 10V;
(this can be used worldwide,i will send the the correct one to fit
E). Frequency of electrical source: 50Hz± 1Hz.
Surroundings of transportation and reservation
A). Surroundings temperature range: -40o C-+55o C;
B). Relative humidity range: ≤93%
C). Atmospheric pressure range: 500hPa-1060hPa.
(1). Processing time of film
The time which develops films needed is 2min-8min, and can be
accommodated continuously. The error is within± 10%.
(2). Light leaking
During the bathing process, light leaking is not permitted.
(3)The stability of film delivery
The delivery mechanism of film should not lacerate or lock films.
(4)The timing panel of transit film
The timing panel of transit film adopts the number 1-12 represents
month, each scale represents 10 days.