The HN-05 oral automatic X-ray film bath equipment is a total
automatic and wall mounted type. The film is bathed automatically
through three step: Developing, fixing and bathing. The time
developing films needed can be selected in the designing range.
During the bathing process, delivery mechanism adopts anti-flaming
ABS engineering materials. It has easy cleanliness and maintenance,
stable running, small volume, operators do not touch the developers
and fixers directly, convenient and safe operation and so on
2. Main Technical parameters
2.1Normal working conditions
A). Surroundings temperature: 10o C~30o C;
B). Relative humidity: ‰¤ 70%;
C). Atmospheric pressure: 700hPa~1060hPa;
D). Voltage of electrical source: 220V± 10V;
E). Frequency of electrical source: 50Hz± 1Hz.
2.2 Surroundings of transportation and reservation
A). Surroundings temperature range: -40o C~+55o C;
B). Relative humidity range: ‰¤ 93%
C). Atmospheric pressure range: 500hPa~1060hPa.
(1). Processing time of film
The time which develops films needed is 2min~8min, and can be
accommodated continuously. The error is within± 10%.
(2). Light leaking
During the bathing process, light leaking is not permitted.
(3)The stability of film delivery
The delivery mechanism of film should not lacerate or lock films.
(4)The timing panel of transit film
The timing panel of transit film adopts the number 1~12 represents
month, each scale represents 10 days.
2.4The product structure
It is composed of Main body , Light-tight box, Delivery
mechanism of film, Liquid cup and Screen and so on.
2.5Input power: 4VA.
It accords with electric equipment safety regulation of the B type
in II species in the national standards GB9706.1.&
X-ray Film Processor (LK-C31) Images